110/220V Acupuntura Pen Point Detector Electronic Acupuntura Massage Pain Therapy Electric Acupuncture Meridian Pen
1. Based on Traditional Chinese Medicine Theory, does no harm to human health
2. The NO.1 inteligent acupoint detection device of the Acupuncture and moxibustion therapy device industry,everyone can be a good TCM doctor at home
3. Iso13485 and CE approved
4. Utillity model patent
5. Appearance patent: CE Digital LCD display electric acupuncture point pen
1. Warming and activating meridian
2. Promoting blood circulation by removing blood stasis
3. Relieve pain,improve immunity
4. Regularize internal secretion
How to use this acupoint pen?
Product Operation Steps:
1. "ON/OFF" knob to around 4 files, then "Sensitivity" spin around to 2, about 2 per second when detecting acoustic hum shall prevail.Due to personal skin and dry humidity weather, "Sensitivity" regulation vary from person to person.
2. Before use, first will need to use in the treatment of probe is installed in the host, and then insert the conductive rod wire in the host at the bottom of the interface, another hand hold the contact rod form a loop.
3. Correct posture of hand instruments, such as holding pen, thumb contact instrument metal conduction piece 1, escaping contact conduction metal piece 2, erect hold in contact with human skin.
4. Detection: the meridian entry points in detecting meridian acupoints, when they hear a continuous hum and display values increase rapidly, the channels and collaterals acupuncture points are detected, then press start treatment under treatment.Note: when detecting meridian acupoints, may hold the key treatment。
5. Treatment, can through the "ON/OFF" key to adjust the intensity of appropriate treatment, in order to feel comfortable it is advisable to (a total of 10 strength level is adjustable from 0 to 10)。
6. After using, press the "ON/OFF" key to adjust to the closed position.
7. Replace beauty probe: beauty face, remove the probe counterclockwise, then clockwise lock beauty probe.
8. When charging, liquid crystal display: with the words“ CH6 ”and flashing, the green light, fully charged, the green light.
Output Intensity 0 to 10 mA
Output Intensity Control ON / OFF, 0 to 10
Power Supply 9 volt battery
Micro-current Output 0 to 2 microamps
Sensitivity Adjust Control 0 to 10
Frequency Control 1 to 16 Hz
Large LCD Digital Display Frequency / Sensitivity Indicator
Indicator Lamp Pulse Rate Indicator
Pulse Width 220 microseconds
Pulse Modes Continuous
Pulse Shape Asymmetric biphasic square wave
Point Detection Meter, Sound and Light Indicators
Output Channels One
Box Size 290 x 180 x 60 mm
1. Do not use for head
if have cardiac conditions and/or pacemakers ,lung device or others medical electronic equipment in the body do not use this acupuncture device at the same time
2. do not use it during pregnancy , menstrual periods , abnormal skin ,fever .
3. do not give no expression ability people to use this acupuncture device
4. be sure to keep out of the reach of childrenAbout battery:
Electric Acupuncture therapy device requires one pc 9V battery. use PP3, 6F22 or its equivalent type of battery (Alkaline battery is recommended). The battery is replaced by opening the battery door on the bottom side of the Unit. Slide the battery door off, remove the old battery and insert the new battery. Insert the battery correctly with the polarity according to the diagram shown in the battery compartment label. Make sure the directions of positive and negative poles of battery inserted is correct and be sure you are using a fresh battery.
Battery life is 4 to 20 treatment hours depending on treatment time used. Many short treatment sessions give longer battery life than fewer longer sessions. to avoid battery leakage, remove the battery if the unit will not be used for an extended period of time.Cleaning and maintainence DO NOT IMMERSE THE Electric Acupuncture therapy device II STIMULATOR/DETECTOR IN ANY CLEANING SOLUTION.
1. Should periodically wiped clean the unit using a damp cloth or neutral detergent.
2. The metal electrodes should be thoroughly washed in a 50% mixture of isopropyl alcohol and water.
3. The wire should be wiped clean with a damp cloth or neutral detergent and then wiped dry.
4. To avoid cross infection suggest use Alcohol to cleaning the equipment probe after used.
5. Electro Pad (Ashi pad) stick surface should be keep cleaning .after use should be stick back on the plastic board and packing them immediately.
6. Can use water to cleaning the stick surface to resume the viscous. if the viscous not enough should be change a new one.